03964.com

文档资料库 文档搜索专家

文档资料库 文档搜索专家

Bias Stability measurement

Bias Stability Measurement: Allan Variance

Dr. Walter Stockwell Crossbow Technology, Inc. http://www.xbow.com The Crossbow Technology 700 series vertical gyros (VG) and inertial measurement units (IMU) have a specification for bias stability at constant temperature. What does bias stability mean? Figure 1 shows a typical output of rate vs. time for a VG700CA unit. Note that the peak-to-peak noise is about 0.38 deg/sec or about 1400 deg/hr. How can we measure a bias of less than 20 deg/hr? Bias is a long term average of the data. It is meaningless in terms of a single data point. To measure the bias, we must first take a long sequence of data, and find the average value. Clearly, in Fig 1, the bias is about 0.15 deg/sec. Bias stability refers to changes in the bias measurement. For example, what would the bias be if we took data two hours from now? To measure bias stability, we need to measure the bias many times and see how the bias changes over time. Even this leaves some open questions: for how long should we average the data; how many times should we measure the bias to make a valid measurement of the bias stability?

Figure 1. VG700CA rate output Fortunately, all of these questions have been worked out already. The Global Positioning System (GPS) depends on accurate time measurements, and clock stability is an important parameter in determining the ultimate accuracy of GPS. Dr. David Allan worked out a method of characterizing noise and stability in clock systems – this method is now called the “Allan Variance.” The Allan Variance (AVAR) is a method of analyzing a time sequence to pull out the intrinsic noise in the system as a function of the averaging time. It was developed for clocks, but we can easily adapt it for any other type of output we are interested in. Here is the basic idea. Take a long sequence of data and divide it into bins based on an averaging time, τ. Average the data in each bin. Now take the difference in average between successive bins, square this

Page 1

Bias Stability measurement

number, add them all up, and divide by a rescaling factor. Take the square root of the result. You now have a quantitative measure of how much the average value changed at that particular value of averaging time. Go back and increase τ, and start over. Keep doing this until you get to about 9 time bins to average over – any less, and the results begin to lose their significance. See figures 2a, 2b, and 2c. Here is the actual equation:

AVAR2 (τ ) =

1 ( y (τ ) i+1 ? y(τ ) i )2 , ∑ 2 ? ( n ? 1) i

where AVAR( τ) is the Allan Variance as a function of the averaging time, τ; y i is the average value of the measurement in bin i; and n is the total number of bins. When you are all done, you can graph the AVAR results as a function of τ. Figure 3 shows the results for this data, along with the error in the calculation. For clarity, the AVAR data is plotted on a log-log scale.

Figure 2a. Data divided into bins of 1 second length

Page 2

Bias Stability measurement

Figure 2b. Data divided into bins of 10 second length

Figure 2c. Data divided into bins of 500 second length

Page 3

Bias Stability measurement

Figure 3. The Allan Variance result for the rate sensor data. (The red lines show the error limits in the calculation) At short averaging times, the Allan Variance is dominated by the noise in the sensor. There is a direct correlation between the standard deviation (the noise) of the output vs. time with the slope of the Allan Variance at small τ. This is also referred to as angle random walk (ARW.) As you average over longer and longer times, the variance decreases. This makes sense if you compare Figures 2a and 2b. As you increase the averaging time, you get a better measure of the bias, and therefore would expect a subsequent measure of the bias to be close to your first measurement. An interesting thing happens as the averaging time increases. At some point the Allan Variance starts to increase again. This is due to rate random walk (RRW) in the sensor – inherent instability in the output of the sensor. The standard definition of bias instability used by inertial sensor manufacturers is the minimum point on the Allan Variance curve. This is the best stability you could achieve with a fully modeled sensor and active bias estimation. The sensors used in the 700 series IMUs and VGs will typically be within 20 deg/hr bias stability with a 10 – 15 sec averaging time.

References

Crossbow Technology, Inc. http://www.xbow.com ? See all of Crossbow Technology’s products: inertial sensors, accelerometers, magnetometers, and cutting edge wireless networked sensors. Download product data sheets, manuals and software.

Page 4

Bias Stability measurement

Dr. David Allan’s website about Allan Variance Calculations. http://www.allanstime.com/AllanVariance/index.html ? This website is a good place to start in understanding the Allan Variance. He has equations, a good explanation, and links to applications and technical discussions.

IEEE Std 952-1997, "Guide and Test Procedure for Single Axis Interferometric Fiber Optic Gyros," IEEE, 1997, p.63. IEEE standard for defining methods to characterize FOG sensors. Includes a long appendix describing the Allan Variance measurement in more detail.

Page 5

相关文章:

- 3D Gyro.doc
- 3D
*Gyro*- 3D*Gyro*(三轴陀螺仪) 2010-12-29 14:

- gyro_图文.pdf
- /s. One practical application is to measure how quickly a car turns by mounting a
*gyro*inside the vehicle; if the*gyro*senses that the car is ...

- 船舶有关词汇.doc
- 询问机 echo-sound 回声测深仪 radio navigation equipment 无线电导航设备 log 计程仪 auto-pilot 自动驾驶仪
*gyro*-compass 电罗经 barrier 屏障,障碍,栅栏 zenner...

- 3D Gyro_图文.pdf
- 3D
*Gyro*- 豆瓣社区 豆瓣读书 豆瓣电影 豆瓣音乐 豆瓣同城 九点 豆瓣

- 6050程序.txt
- MPU6050_CLOCK_PLL_X
*GYRO*); 104 MPU6050_WriteBits(MPU6050_DEFAULT_ADDRESS, MPU6050_RA_*GYRO*_CONFIG, 105 MPU6050_GCONFIG_FS_SEL_BIT, MPU6050_GCONFIG_...

- 陀螺定向测量.doc
- 陀螺定向测量 - 陀螺定向测量 陀螺定向测量(
*gyro*static orient

- 车模直立方案_图文.pdf
- 第9页 互补滤波-一种很好的控制思想 IIC 移植完毕后直接调用函数,就可以采集到 X 与 Y 方向的角速度值了:
*GYRO*_X=Get_*Gyro*(1,'X')30; //用于...

- 改进的单轴旋转SINS初始对准方法.pdf
- (NavalAeronauticalEngineeringInstitute,Yantai264001,China) WANGDong,LIGuolin Abstract:Focusingontheissuethatthelimitofaccuracyisrestrictedbytheaccuracyofthe
*gyro*and...

- 沙特阿美公司钻修井工艺流程简介--李艳民_图文.ppt
- (g)下入的钻具按
*GYRO*的要求进行通径,盖好井口。 (h)严格控制下钻速度(1

- Data-driven fault diagnosis for an automobile suspension ....doc
- ?A
*gyro*meter measures the pitch and the

- 1999-(马铁华)-地磁场传感器及其在飞行体姿态测量中的应用.pdf
- D esign of a so lid 2sta te
*gyro*scop ic

- GyroCalibration_EDN_EU_7_2010.pdf
*Gyro*Calibration_EDN_EU_7_2010 - SILICON-

- datasheet_sar150_high_precision_gyro_sensor_ts1514_r1.pdf
- datasheet_sar150_high_precision_
*gyro*_sensor_ts1514_r1 - DATASHEET SAR150*GYRO*SENSOR SAR150 (Sc...

- 导航学(第一章)导航系统概述_图文.ppt
- 1852年发现了陀螺效应,法国科学家首先使用 “
*Gyro*”(Gyroscope

- 米切尔 MICHELL ENGINEERING GYRO SE LP唱盘_论文.pdf
- 米切尔 MICHELL ENGINEERING
*GYRO*SE LP唱盘 - 来

- MWC_computeIMU算法解析(修改)20130908.doc
- It works with 3ms //
*gyro*only: the delay

- GYROSHOT.unkown
*GYRO*SHOT precision borehole surveying technologies*GYRO*SHOT Gyroscopically-oriented Borehole Survey Tools EASY TO USE*Gyro*Shot tools are ...

- P-494 Gyrokinetic.unkown
- (ITB), have been examined with
*gyro*kinetic simulations with the GS2[1,2] and*GYRO*[3] codes to identify the underlying key plasma parameters for ...

- itg3205 gyro datasheet.unkown
- Tel: +1 (408) 988-7339 Fax: +1 (408) 988

- CS-GYRO Report Summary.unkown
- CS-
*GYRO*Report Summary Proje ct re fe re nce : 271494 Funde d unde r: FP7-JTI Final Repo rt Summary - CS-*GYRO*(MEMS*gyro*meter fo r wing ...

更多相关标签:

- Measurement Bias
- On bias plus variance
- UNIVERSITY OF CALGARY Modeling Inertial Sensors Errors Using Allan Variance
- Estimating bias and variance from data
- The Bias Variance Tradeoff
- Bias, variance and prediction error for
- Bias and Variance of Rotation-based Ensembles
- Performance Improvement of GPS_INS Integrated System Using Allan Variance Analysis
- Conceptual complexity and the bias-variance tradeoff
- Minimal-Drift Heading Measurement using a MEMS Gyro for Indoor Mobile Robots